Accutrend GC Blood Pressure Monitor pdf manual download. Since each code strip and test Glucose ® Accutrend, GC, GCT, DM CODE 50 TESTS. Description Reviews (0). ACCUTREND GCT PLUS BLOOD ANALYSIS METER. Test Report For: Glucose User Manual. Optional Request: In Exchange For. DownloadAccutrend gct user manual. Semoga Download driver keychain coby dp usb win xp, Hi osvaldorodriguez, are you If you are still a Windows user.
|Published (Last):||21 April 2005|
|PDF File Size:||6.87 Mb|
|ePub File Size:||17.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
Accutrend GCT System is a combo monitoring unit for cholesterol, blood glucose yct triglycerides, all in a fast, easy and reliable measurement process. This is one major self-testing device to put you in good control of the direction of your health, even as you strive for the accomplishments you desire.
If at any time during the function test mnual display does not correspond to either of the diagrams on the left, please notify the Roche Diagnostics service see pages 97 and A performance check should be carried out: Don’t show me this message again. Plausibility check 37, 69 Practising measurement 29ff, 59 Retrieval of results from memory 49ff.
Ready for setting avcutrend date and time or the beeper. They are essential to ensure precision and reliability of the measurements. I like what you guys are doing just keep it up! The Accutrend GCT System is most suitable for accurate on-site screening, diagnosis, and therapy monitoring of metabolic disorders and cardiovascular risk factors. If the problem persists, contact the Roche Diagnostics service centre see Section 6. Safety accutrrnd such as are also stored. Page of 58 Go.
Storage conditions Switching on Symbols 4, 5 Technical data Test procedure 29ff. Page 36 If the result is again outside the specified range, consult your nearest Ch. If other messages appear on the display, consult Sec- Too little blood applied — With the GCT System, you easily do these tests kanual your own, at your convenience, with accurate, dependable results.
Roche Accutrend GC User Manual
Try a Test-drive for Yourself Now! The test strip is not suitable for the meter. Then CODE stops flashing. Blood collectionTiong Bahru Road Measure three important health parameters – cholesterol, blood glucose and triglycerides in one easy process, leaving you in better control of accuternd health. The time must also be changed from summer to manal ter time and vice versa, and 29 February must be entered in a leap year. Code the meter see Section 3.
Humidity In the presence of high humidity or large temperature changes condensation may form on the meter, e. After lancing the finger remove the test strip from the open meter and place on a piece of clean, absorbent material, e. By pressing ‘print’ button you will print only current page.
The test strip is too old or has been incorrectly stored.
If you suspect a fault in your meter you should first contact the advisory service at your nearest Roche Diagnostics office. Light from an LED light emitting diode is directed accutrene the test strip from below and the blank value of the test strip determined by measurement of the reflected light.
ROCHE ACCUTREND GC USER MANUAL Pdf Download.
Press the SET button briefly. Don’t have an account? If the button is released earlier the clearing process will be interrupted.
The round window on the back of the strip is not uniformly coloured. If too little blood is applied, do not try to spread it or apply a second drop as this may also produce incorrect results.